9110 Table Top Dispenser
The 9110 is a "plug and play" table top dispenser system for laboratory and dosing applications. It is supplied complete with pump unit, 1m of tubing, dispenser tip and holder, set of tube clips, universal power supply and instruction manual.
Fitted with a quality 24 V DC motor geared to either 320 or 520 rpm the 9110 is a fully programmable, compact, and ecconomic dispenser.
› Simple calibration to tube size and media (reagent)
› Dose to exact required volume
› Time interval dispense mode (programmable delay and number of repeats)
› Electronic flow control to 180 ml/min
› Continuous dispense mode
› Input for remote foot pedal (or system signal)
› Flow direction switch
› 24 V DC for safe operation
› Easy and quick tube replacement
› Compact construction
› Self priming
› Range of alternative tube materials and sizes available.
Return to Peristaltic Pumps for Liquid range.
Available to purchase online via BOX-it (see order information below).
Although shown here as a finished system the 9110 can additionally be customised to meet OEM requirements.
|Flow Data (320 rpm model, part number 9110.000)|
|ID Ø 1.0 mm tube||to 15 ml/m|
|ID Ø 2.0 mm tube (as supplied)||to 50 ml/m|
|ID Ø 3.0 mm tube||to 110 ml/m|
|Flow Data (520 rpm model, part number 9110.010)|
|ID Ø 1.0 mm tube||to 25 ml/m|
|ID Ø 2.0 mm tube (as supplied)||to 85 ml/m|
|ID Ø 3.0 mm tube||to 180 ml/m|
|Tube Materials||Silicone (as supplied) / Lagoprene / Pharm-a-line / ED-Plex|
|Voltage||100 to 240 V / 50 to 60 Hz|
|Power Supply Unit Output||24 V DC 1A|
|Adapters||EU, UK, USA|
|Max pressure||1 bar|
|Max vacuum||-950 mbar|
|DC motor life||>2000 hour|
|Operating Temperature Range||10 to 40 °C|
|Storage Temperature Range||4 to 40 °C|
|Operating Relative Humidity||90%|
|Storage Relative Humidity||95%|
|CE marked to EEC directives|
6013.000 Foot Switch
9100.005 Dispenser Tips 1-100µl (x 25)
9100.001 Mounting bracket / stand
9100.006 Dispenser set (1m tubing, connector, and dispenser tip)
3022.000 8 Way Dispenser Comb
All data is representative for initial selection purposes. It is the responsibility of the user to determine suitability for the intended use. These products are not suitable for in-vivo applications.